SNC Machine™ listens for and captures your QA files, processes and analyzes the files, and saves the results to the database. Simply log in to the application and immediately view a dashboard of results. Accept results that pass, and drill down into the analysis details for results that fail. Trend any piece of data against any other piece of data. It is that simple, and that powerful.
All image-based tests for TG-142 imaging and mechanical QA, plus VMAT QA, are included with SNC Machine. Simply deliver the test beam and SNC Machine does the rest. Accept or reject results on your terms and timeframe.
All VMAT tests recommended by Varian are included with SNC Machine.
19 different tests for TG-142 & VMAT QA
The SNC Machine Dashboard presents all recently executed tests. Failed tests are highlighted red. Approve the tests from the dashboard or click the test to drill down and scrutinize the results.
In this MLC Picket Fence test, failed leaves are highlighted, as are the pickets that failed.
Measure TG-51 protocols in pre-configured workflow and capture Winston-Lutz Isocenter Tests that are automatically calculated, supporting your less routine Annual QA machine tests.
Results and images for a Winston-Lutz Isocenter Test. View the 2D results for each image as well as the combined 3D results of all images.
Create trend graphs from any test result and easily compare trends using a single graph.
Results and images for a CBCT Test. Rollover any thumb nail to see the parameters and the specific corresponding ROI’s to ensure image registration was performed correctly.
Trend and compare the test parameter against other test parameters from any number of machines. Easy to read graphs display trend lines relative to custom tolerances.
|TG-142 Imaging:||CBCT Image Quality, kV Image Quality, MV Image Quality|
|TG-142 Mechanical:||MLC Picket Fence, MLC Positioning & Leaf Speed, Winston-Lutz Radiation & Machine Isocenter, Gantry/Couch/Collimator Starshot, Light/Radiation Congruence|
|Beam:||Field Size, Beam Flatness, Beam Symmetry|
|VMAT:||Dose Rate versus Gantry Speed, Leaf Speed, Arc Point Dose, DMLC Point Dose|
|Sun Nuclear:||MV-QA, kV-QA, FS-QA, WL-QA|
|Standard Imaging:||PipsPro Phantoms|
|Phantom Laboratory:||CatPhan 503, 504, 600, 604|
|Varian:||Las Vegas Phantom|
|PTW:||EPID QC Phantom|
|Dimensions:||60 x 60 x 60 mm|
|Sphere Size:||7.0 mm|
|Sphere center accuracy:||.2 mm|
|Field sizes:||100 x 100 mm; 150 x 150 mm|
|Markers (±0.1mm):||56 - Field siz e (7 per field edge) 1 - Orientation|
|Dimensions: (L x W x D):||178 x 178 x 6 mm|
MV-QA / kV-QA
|MV Line pairs:||0.1, 0.2, 0.5, 1.0 ± 0.025 mm|
|kV Line pairs:||0.6, 1.2, 1.8, 2.4 ± 0.01 mm|
|MV ROI:||9 (4 spatial, 4 contrast, 1 center)|
|kV ROI:||28 (4 spatial, 23 contrast, 1 center)|
|MV Dimensions: (L x W x D)||127 x 102 x 25 mm|
|kV Dimensions: (L x W x D)||127 x 127 x 16 mm|